However, for maximum portability, the instrument address should use uppercase characters only. Now for the Newport device, you want to place the folder that is contained in the zip file that is downloaded into the C: All of directory paths which follow assume Windows No all statistics defined in the help file were returned. This error effected the range tests in the following function: Message 9 of This command will be fixed in a future release of the firmware.
|Date Added:||4 May 2011|
|File Size:||47.13 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
This error effected the range tests in the following function: There are several advantages to using the dll form of the driver. You will see a agx sub-pallette. Message 9 of The AG X driver is located in a self-extracting archive file called X.
Do I need to initialize, configure and close the instrment driver? All of directory paths which follow assume Windows Does anyone know what happens? The driver uses Pascal calling conventions. Units were modified from m to nm and Hz to GHz. The message shows “An exception occurred within the external code called by a Call Library Node. If you still get the same errors it would be best to contract Agilent because the driver is not certified or maintained by National Instruments but by Agilent.
No all statistics defined in the help file were returned.
Instrument Control (GPIB, Serial, VISA, IVI)
Message 6 of Message 10 of Message 3 of I can use other library VI of the driver to change the centre wavelength and span. Do not build the driver in the directory in which it was installed. agileht
It includes a comprehensive on-line help file which complements the instrument manual. I am using Labview 7. I am a Labview begineer and really don’t know how can I solve the problems. Copy the files in the C: It includes a Visual Basic include file.
Message 7 of Please consult the documentation before using functions in the new driver. Vierror occurrs. The help file presents application programming examples, a cross-reference between instrument commands and driver functions, and detailed documentation of each function, with examples.
Agilent | Agilent x VXIplug&play Instrument Driver Read Me
Transportability across different computer platforms 2. To use source code written for a previous driver, the older driver must be installed and its. The interface 886142b some functions in this driver have changed from similar functions with a different prefix in older versions of the driver.
The help file is in error. By placing these files in that folder in the instr. Create a folder agx in the C: I have also downloaded the instrument driver from NI 861422b.
Added 20 new system and application functions.